- Data distribution
- Using the Archive
- Gemini Home
- Telescopes and Sites
- Science Visitors at Gemini
- Observing With Gemini
- Retired Instruments
- Visiting Instrument Policy
- Visiting Instrument Telescope Interfaces
- DSSI Speckle Camera
- TEXES (North)
- Integration Time Calculators
- Magnitudes and Fluxes
- Near-IR Resources
- Mid-IR Resources
- Observing Condition Constraints
- Performance Monitoring
- SV/Demo Science
- Future Instrumentation & Current Development
- Queue and Schedules
- Data and Results
- Calibration Data Retrieval
- Past/Future Development
Change page style:
GNIRS Imaging (with sensitivity table)
GNIRS can be used for near-infrared (1.0-2.5µm) imaging science and is in fact quite sensitive. However, as a photometer/imager it has the following limitations.
- The order-blocking filters have non-standard bandpasses. Only the H order-blocking filter approximates the Mauna Kea photometric filter for its band.
- For the order blocking and narrow band filters the field of of view is the acquisition "keyhole," whose dimensions are shown below. For the Y, J, and K Mauna Kea filters the un-vignetted field is much smaller, as also shown (the Observing Tool does not yet show this smaller field).
- Currently the best image quality that has been achieved by GNIRS + adaptive optics (ALTAIR) is ~0.12 arcsec (fwhm); this is twice the diffraction-limited fwhm in the K band.
Observing proposals that involve infrared imaging should request NIRI, not GNIRS. If it is not known whether NIRI will be available, Phase 2 may involve creating observations for both instruments. GNIRS also may be used to perform rapid Target of Opportunity (ToO) imaging programs if NIRI is unavailable. Prospective users should be aware that the Gemini IRAF package contains no dedicated support for GNIRS imaging reductions.
The basic detector properties are given here. See also the known issues page. Approximate sensitivities and maximum exposure times (shallow well, except where noted) for imaging though several filters in GNIRS are given in the table below. Sensitivity values are for 0.15" pixels without AO; they are expected to be similar for 0.05" pixels with AO. The Y filter is not yet ready for use.
|GNIRS IMAGING SENSITIVITY IN GOOD SEEING AND WITH 0.15" PIXELS
(S/N=5 IN 1 HOUR, N.I. OVERHEADS)
|Filter||Bandpass (µm)||Point Source magnitude||maximum
|Y (Mauna Kea)||0.97-1.07||tbd||tbd|
|X (order blocking)||1.03-1.17||23.9||50 s|
|J (Mauna Kea)||1.17-1.33||23.5||90 s|
|J (order blocking)||1.17-1.37||23.7||50 s|
|H (order blocking and Mauna Kea)||1.49-1.80||23.0||10 s|
|K (Mauna Kea)||2.03-2.37||22.6||30 s|
|K (order blocking)||1.90-2.49||23.1||10 s|
|PAH||3.27-3.32||15.7||0.8 s (deep well)|