- Gemini Home
- Telescopes and Sites
- Science Visitors at Gemini
- Observing With Gemini
- Retired Instruments
- Visiting Instrument Policy
- Visiting Instrument Telescope Interfaces
- DSSI Speckle Camera
- TEXES (North)
- Integration Time Calculators
- Magnitudes and Fluxes
- Near-IR Resources
- Mid-IR Resources
- Observing Condition Constraints
- Performance Monitoring
- SV/Demo Science
- Future Instrumentation & Current Development
- Queue and Schedules
- Data and Results
- Gemini Research Staff
Change page style:
Michelle spectral flats
28 Feb 2005
It has recently been determined that all spectral flats using the low resolution 10um grating obtained prior to Feb 27, 2005 were saturated. This means that:
- all unratioed spectra that have been flatfielded prior to then have incorrect spectral shapes;
- all spectra of extended sources that were ratioed by spectra of a point source (std) have incorrect shapes.
These differences are not huge, but are noticeable.
Note that reduced spectra for which the spectrum of a point source target was ratioed by the spectrum of a point source standard, both measured in the same or nearly the same array rows, are unaffected and OK.
If you have old Michelle spectral data and would like an unsaturated flat to use in re-reducing those data, please contact Tom Geballe or Scott Fisher.
Last update Feb 28, 2005; Tom Geballe and R. S. Fisher